New features in the volumetric scanning panel allows users to quickly populate the ETL table to perform measurements at equally spaced points. While scanning, simply move the ETL to the top of the volume you want to image and click the “Top” button to copy the location. Do the same for the bottom limit.
Once the limits are set, you can stop scanning and select the step size (in um) and the number of frames you want the system to scan at each location. As you change the parameters, the table will automatically reflect those changes.
Click the “Enable” checkbox to upload the table to the Scanbox tower and enable volumetric imaging. Wait for the table to finish uploading. Now, when you scan, the system will scan the volume as requested.
An example of the process is shown below: